Thermocouple Welder by DCC Corporation
The original HotSpot solves your basic thermocouple fabrication and attachment problems. It generates an electric arc for fusing standard thermocouple wires into freestanding beads and can weld them to any thickness base metal.
In The Box HotSpot I Portable Welder (5 to 50 watt/sec) HotSpot I Welder and Metal Case, Standard 6in Wire Holding Pliers, Eye Shielding Protective Goggles, Carbon Block, Grounding Magnet, Instruction Manual.Compare with Other Models
CONTROLS, INDICATORS, & OVERLOAD PROTECTION
Visual and sonic indicators alert operator when the welder has cycled and is at the user selected power level. The system is powered directly from the AC line through a step-down transformer, or from a gel type battery, which provides power for a few hundred welds off-line. The battery is automatically recharged whenever the unit is connected to AC power.
The power level available for welding is set by the position of the front control knob. This control also functions as the on/off switch. The knob is calibrated in Watt seconds or Joules. The maximum available power is approximately 50-Watt seconds. The control actually regulates the voltage to which the energy storage capacitor is charged. The stored energy is proportional to the square of the voltage, and the voltage can be varied between 15V and 80V. Increasing the setting of the control knob will cause the capacitor to be charged to the higher level. However, decreasing the setting will not reduce the value already stored, so a welding cycle will always release an energy pulse equal to the highest setting since last recent discharge.
When the internal circuitry has charged the energy storage capacitor to the level set by the control knob, the front panel LED and internal sounder are activated. These inform the operator that the unit is ready for another cycle. The sounder also helps to conserve battery energy during portable operation by reminding the operator that the welder had been left on.
A circuit breaker protects against overloads. The circuit breaker opens the low voltage AC/DC supply if excessive current is being drawn. A 1/4 Amp fuse mounted on the internal circuit board is in series with the AC power cord.